The detector is intended for the detection and visual observation of particulate fluxes and electromagnetic radiation as a part of scientific and industrial equipment.
The detector consists of a chevron stack of two MCPs, a phosphor screen and ceramic-to-metal fittings. Its operating principle relies on the detection and amplification of the primary flux of charged particles with subsequent imaging of the amplified flux on the phosphor screen.
The detector can be vacuum heated at 300°C. The detector can be operated in any position.
|MCP active area diameter||mm||48|
|Single electron pulse height resolution, max||%||120|
|Dark count rate density||count/sec×cm2||max 3|
|Pressure when in operation||Pa (Torr)||1.3×10-4 (1×10-6)|